A 938Kb PDF of this article as it appeared in the magazine—complete with images—is available by clicking HERE

Note: Pete Yates-Hodshon of Tucson pointed out that the notation of the formulas are incorrect. Here's what Shawn had to say:

Please Excuse My Dear Aunt Sally or Parenthesis Exponents Multiplication Division Addition Subtraction. It needed the extra brackets to make sure the division is done first instead of the multiplication.

The formulas below have been corrected, but the formulas in the PDF cannot be corrected.

Also note that two more hidden point offset techniques can be found HERE

Let's say it's 5:30 Friday afternoon and you're past ready to call it a week. You've just calculated the angle and distance to look for one of the last monuments you need to tie in. You turn the instrument to the angle and shoot a distance that measures just behind a tall tree. After a few minutes' search you find the monument intact and, sure enough, it is behind the tree. You're glad the point exists, but irked because you can't see it from the instrument.

Your brain begins to click through a mental toolbox of offset techniques. There are still two more setups and a delay like this is not on the agenda. But wait! Before you add another traverse point (or cut the tree down), there is another alternative the hidden point offset.

For me, the hidden point offset has always brought to mind an old instrument ad that showed a pole with two prisms on it. The idea was to shoot the two prisms, after which a special software routine would calculate the coordinates at the base of the pole. It seemed like a nice idea, but I was skeptical about the accuracy. Add to that the cost of special hardware and software, as well as the rare occasion that would necessitate such a tool, and such a method just seemed beyond the reach of practicality.

One of my favorite aspects of surveying is problem solving. Like many surveyors, I enjoy unique situations that require creative uses of mathematics and mechanics. A couple of years ago I began to ponder whether I could do an offset with normal tools. As I thought about the mathematics involved and formulated a method, I was pleased to discover that the math could all be worked out with a simple, non-scientific calculator, like one on a cell phone. No special equipment was necessary, either. As long as a rodman can set up the prism pole on a point without the pole shifting or slipping, the technique can be performed with a common prism and prism pole.

Try This!
Here's how it works. Standing behind the tree that is blocking the view, the rodman centers the pole on the apex or center of the point and leans the pole over such that the prism can be seen from the instrument. The pole will need to be locked firmly in place during the procedure, which will likely require a solid bipod. Next, the instrument operator shoots the prism with the pole lowered as low as possible. If a data collector is being used, the rod height should be set to zero and the actual rod height should be carefully noted in a field book or in the description of the shot. Next the pole should be raised as high as is practical and a second shot made. Again, if the shot is being recorded in a data collector, the rod height should be set to zero and the actual rod height should be recorded in a field book or in the shot description. The pole should be calibrated to read true rod heights from pole point to prism.

Once the coordinates of the two shots (low and high) have been determined, a simple formula can be used to determine the coordinates of prism pole point:

Northing
Nl - [(Nl - Nh)/(HRl - HRh)] * HRl = Np

Easting
El - [(El - Eh)/(HRl - HRh)] * HRl = Ep

Elevation
Zl - [(Zl - Zh)/(HRl - HRh)] * HRl = Zp

Nl Northing coordinate of the short pole shot
Nh Northing coordinate of the long pole shot
Np Northing coordinate of the actual point

El Easting coordinate of the short pole shot
Eh Easting coordinate of the long pole shot
Ep Easting coordinate of the actual point

Zl Elevation of the short pole shot
Zh Elevation of the long pole shot
Zp Elevation of the actual point

HRl rod height of short pole shot
HRh rod height of long pole shot

This procedure requires no special equipment or special poles and is actually more adaptable than using the special double-prism pole because you have the flexibility to fine-tune a specific height for your particular needs (such as using 5.2 for the low shot to get over the wood fence or 7.8 for the high shot to stay under a tree limb).

Perhaps most pleasantly surprising is the accuracy I've been able to achieve with this technique. However, it must be pointed out that this isn't a panacea and care should be taken when performing a hidden point offset. Error sources beyond the normal surveying sideshot error sources are: 1) movement of the pole alignment between shots, which causes you to unknowingly lose your vector along the pole; 2) incorrect rod heights; 3) relatively low ratio of distance between high and low shots and the low shot and pointyou don't want to project a half foot out five feet; 4) misalignment of the prism leading to parallax.

Regarding the calculation, most companies probably don't want their field crews doing algebra in the field. Fortunately, both AutoCAD and IntelliCAD have a routine called LENGTHEN which can perform this operation graphically back in the office.

Simply draw the 3D line between the high rod shot and the low rod shot and lengthen by the low rod height. It can be done in just a couple of seconds. Also, for surveyors who are also programmers, if your data collector will allow it, it would take little effort to write a small program that could perform this in the field. Additionally, it isn't too difficult to perform the calculations manually using paper, pencil and a basic calculator.

As to the issue of parallax, our friends at SECO sent us a true nodal prism to try out for this article, but you don't need a true nodal prism. If you are careful to point the prism to the instrument, parallax should not be an issue. However, at a rod height of eight feet or more, it becomes difficult to assure accurate alignment of the prism to the instrument. A true nodal prism places the apex of the prism directly over the center of the rod, meaning that regardless of orientation, vertically and horizontally, the apex, or nodal point, represents the actual center of the rod. With the pole leaning, the normal aids for collimating the prism (center of the pole and prism targets) don't really work. Having a precise, accurate target point will significantly improve the accuracy of the method.

Here are a few examples I observed in the field. For these examples, the instrument was set up at:
N: 5000.0000
E: 5000.0000
Elevation: 100.0000

It was zeroed at a backsight azimuth of 351°12'35". The instrument height was 5.44 feet. For this first example, I set the pole up leaning to the right from the perspective of the pole. I observed at a rod height of 5.00 feet for the low shot and 8.00 feet for the high shot.

At 5.00 feet I observed:
AR: 122°41'18"
ZE: 88°56'51"
SD: 184.962'

This yielded rectangular coordinates of:
N: 4925.0826
E: 5169.0762
Elevation: 108.8382 (Remember to zero the rod height for determining the coordinates.)

At 8.00 feet I observed:
AR: 122°30'57"
ZE: 88°02'08"
SD: 184.950'

This yielded rectangular coordinates of:
N: 4925.6280
E: 5169.2191
Elevation: 111.7807 (Again, remember to zero the rod height for determining the coordinates.)

I encourage you to go out and spend 15 minutes in a parking lot and see what you can do with this method. It is inexpensive to implement and works at times when there just aren't any alternatives. You also get reasonably accurate 3D coordinates from it. As this data set indicates, this method offers verticals that are as precise as the horizontal. And you can amaze your friends at your next outdoor barbeque and say, "Hold my beer and watch this!" In practice, I would encourage you to implement redundancy any time you use this technique. If you are reducing in the field, two separate observations should allow you to compare and make sure the rod didn't shift or you didn't mis-enter a rod height. If you are waiting to reduce it in the office, I would recommend three different observations. This will allow you to pick out the odd duck if you have any disagreement. Enjoy!

Shawn Billings is a licensed land surveyor in East Texas and works for Billings Surveying and Mapping Company, which was established in 1983 by his father, J. D. Billings. Together they perform surveys for boundary retracement, sewer and water infrastructure routes, and land development.

A 938Kb PDF of this article as it appeared in the magazine—complete with images—is available by clicking HERE

Editorial: A Great Year to be a Surveyor
Some magazines have what are called "theme" issues. That is, most of the content is focused on one particular subject. In my 22+ years of survey magazine publishing, my philosophy has always been to have a little bit of everything in each issue, thereby eliminating the possibility that .... Read the Article

Decided Guidance: Case Examinations: Halverson v. Deerwood Village
Whew! We really beat the snot out of Bryant v. Blevins and practical locations. Well this month we're back on new case that hit the Minnesota Supreme Court's docket in 1982. We've got the familiar gymnastics of jurisprudence featuring an extraordinary array of flying rope stretchers ... Read the Article

Extreme Environment Surveying
A Franco-Chilean team of cave divers used the Nautiz X8 rugged handheld for marine cave surveying, gathering data to classify the inaccessible northern half of Madre de Dios for UNESCO World Heritage. The team of cave divers used the Nautiz X8 .... Read the Article

The Original Green Engineers
Sometimes, it's best just to leave things as you found them. That's the lesson shared by Dr. Richard Miksad and his students at the University of Virginia. As a result of studies covering nearly a decade, Miksad's teams have developed detailed .... Read the Article

Test Yourself 49: No Dimensions
In square A-C-D-B with side S, C-E is tangent to the semicircle Q1 with diameter B-D. Q2 is the inscribed circle of A-C-E. The tangent to Q1 and Q2 meets the sides of the square at F and H and intersects C-E at t G. Q3 is the inscribed circle of C-G-H. What is the ratio of the radii of circles .... Read the Article

Discovery on Grizzly Peak
When First Lieutenant Montgomery M. Macomb arrived in Carson City, Nevada, from Washington D.C., on July 28, 1878, his assigned survey crew from the 4th Artillery was waiting and ready for the new field season. At age 25, Macomb was the leader .... Read the Article

Vantage Point: Fighting City Hall Over Land
Once upon a time (1989 to be exact) in a place not so far away from where I live, a man (Francis Galdo) bought a home across the street from a vacant parcel owned by the City of Philadelphia. That parcel, along with others, had been acquired by condemnation back in 1974 subsequent to a 1956 .... Read the Article

Book Review: Boundary Retracement: Processes and Procedures
When I was in my mid-twenties and learning the honorable profession of land surveying, I was lucky to be guided by a mentor who would grab a book off his office shelf and say, "Every surveyor should have a copy of this!" The first example he waved at me was Davis, Foote and Kelly's Surveying .... Read the Article

Reach our audience of Professional land surveyors and Geo-Technology professionals with your GeoJobs career ad. Feel free to contact us if you need additional information.